Is my COM1 port activated at all? Browsing All Articles Articles. Why did you decide not to use the existing driver? I need to create profile, Wirte steps, load profile, Run profile etc. How to assign step numbers in Watlow Profile?
|Date Added:||19 January 2009|
|File Size:||58.35 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. I have tried that. I even don’t know, how to read the code, all I know is to set block diagram from where we can measure things via front panel. Hi All, I am new to this group.
I am new to this group. My first attempt to communicate with this device after installing the driver was by means of the NI MAX. Profile Created will get least available numbers 1 to 40 but step number we need to assign. But in the instrument, the graph is always amplitude versus frequency instead of amplitude vs data point. I’m having trouble making two machines communicate via UDP protocol.
Solved: Query ID for HPA Spectrum Analyzer – Discussion Forums – National Instruments
So i can conclud that my chip actullay connect ok with labview. Channel Catalog Subsection Catalog. See content for details PX statistics example displaying 0. Any thoughts on why this specific VI isn’t working?
But the problem is that it is not using command for amplitude, but for trace A. I am using Query TraceArray. I need to create profile, Wirte steps, 853ec profile, Run profile etc. But I am new to labview, and I am working on it without having a good learning session as I am working in a office. I don’t know realy what to do?
Why did you decide not to use the existing driver? I want to know if their is any way to get the corresponding frequency value too? I have the N chassis and have installed the IVI driver for it.
Instrument Control (GPIB, Serial, VISA, IVI)
It doesn’t like the 0V, the other two options just timeout. How to keep track of step number count? And directly lanview the data to the output by running a loop times. But it seems like it is tracing the data with respect to data points as shown below.
Instrument Control (GPIB, Serial, VISA, IVI) topics
What options do the DSOseries oscilloscopes offer for saving data? In order to enable the control, try reloading the latest cash file by pressing the right button of your mouse and clicking update function label. Please, any help is very welcome. You can save data to internal oscilloscope memory or you can use the NA Scope Connect software to save data to your PC. How can I ensure that my device is properly recognized?
I am trying to connect RS 9 pin via serial ports. Message 3 of 7. I am new to this group.
But I need to collect the frequency value too from the instrument. But it seems like it is tracing the data with respect to data points as shown below. I have two different modules in the chassis.